A facile and high sensitive micro fluorimeter based on light emitting diode and photodiode.
Department of Instrumentation & Analytical Chemistry, CAS Key Laboratory of Separation Sciences for Analytical Chemistry, Dalian Institute of Chemical Physics, Chinese Academy of Sciences, 457 Zhongshan Road, Dalian 116023, China.
Department of Instrumentation & Analytical Chemistry, CAS Key Laboratory of Separation Sciences for Analytical Chemistry, Dalian Institute of Chemical Physics, Chinese Academy of Sciences, 457 Zhongshan Road, Dalian 116023, China. Electronic address: [email protected]
- Published Article
- Publication Date
Dec 01, 2017
A facile and high sensitive micro fluorimeter was developed and evaluated. It employed light emitting diode (LED) as light source, cuvette as detection cell, and photodiode (PD) as optoelectronic detector. Optical and electronic parameters were optimized and demonstrated. A high power LED was chosen, which could irradiate the inner area of the cuvette completely at the same time with divergence angle as small as possible. The optimum LED brought 2.5 times signal-to-noise ratio (SNR) enhancement. Using reflector at the opposite direction of excitation light path doubled SNR. The amplifier circuit of PD was deeply investigated to achieve high sensitivity, low noise, and good stability. The limit of detection (LOD) of fluorescein isothiocyanate (FITC) and chlorophyll at SNR = 3 were 10pM ~ 0.004 ppb and 0.05 ppb, respectively. Basing on the principle structure, a portable fluorimeter for fungimycin detection was developed using a low power UV LED as light source. The LOD for aflatoxin B1 was 0.1 ppb.
Report this publication
From MEDLINE®/PubMed®, a database of the U.S. National Library of Medicine.
This record was last updated on 06/09/2018 and may not reflect the most current and accurate biomedical/scientific data available from NLM.
The corresponding record at NLM can be accessed at https://www.ncbi.nlm.nih.gov/pubmed/28841976